Type of Document Dissertation Author Lee, Kichol Author's Email Address firstname.lastname@example.org URN etd-04192011-143941 Title Effects of Earplug Material, Insertion Depth, and Measurement Technique on Hearing Occlusion Effect Degree PhD Department Industrial and Systems Engineering Advisory Committee
Advisor Name Title Casali, John G. Committee Chair Gauger, Dan Committee Member Keady, John Committee Member Smith-Jackson, Tonya L. Committee Member Winchester, Woodrow W. III Committee Member Keywords
- occlusion effect
- hearing aid
- balloon plug
Date of Defense 2011-01-21 Availability unrestricted AbstractOcclusion effects result from amplification of low frequency components of body- transmitted sound when the ear canal is occluded with hearing protection devices, hearing aids, or other canal-sealing inserts. Since the occlusion effect will enhance the hearing of bodily- generated sounds and result in distorted perception of one‟s own voice, many people report annoyance with hearing aids and hearing protectors that produce occlusion effects. Previous research has studied the effects of ear device insertion depth and influence of the location of the bone vibrator, which has typically been used as the excitation stimulus. However, the effects of monaural vs. binaural, ear device material, and different excitation stimuli were not investigated.
In this research study, the effect of left/right ear canal on the occlusion effect, which was measured objectively as the sound pressure level difference in dB, was investigated. Also, an experiment to determine the effect of earplug types (differing in material and design), insertion depth, and excitation sources was conducted. Lastly, the noise attenuation capability of medical balloon-based earplugs was tested.
Ten subjects, six male and four female, volunteered for the three separate experiments. They were subjected to the three earplug types (foam earplugs, premolded flanged earplugs, and medical balloon-based earplugs), two earplug insertion depth levels of shallow and deep (only feasible with the foam earplug and the balloon-based earplug), and two levels of excitation sources, one of which was a forehead-mounted bone vibrator and the other a self vocal utterance of “EE” to 65 decibels A-weighted (dBA). The attenuation capability of the medical balloon- based earplugs were tested via monaural Real-Ear-Attenuation-at-Threshold (REAT) test per ANSI S3.19-1974 and compared to that of a Peltor H10A earmuff.
Experimental results of the first experiment demonstrated that left right ear canal SPL measurements were not statistically different, and therefore subsequent measurements of occlusion effects for the second experiment were conducted via a monaural left ear measurement protocol. The results of the second experiment confirmed significant effects of insertion depth on the occlusion effect. At the shallow insertion, the occlusion effects, on average, were greater by 11.2 dB(linear) (dBZ) then the deep insertion measured at 500 Hz. The effects of earplug type were mixed. At the shallow insertion, earplug type did not influence the occlusion effect. However, the mean occlusion effect, measured at the 1/3-octave band centered at 500 Hz, of deeply inserted balloon-based earplugs was larger than that of foam earplugs by 3.7 dBZ. Excitation sources that were used as the sound energy stimuli to elicit occlusion effects did not show statistically significant differences. The Noise Reduction Rating (NRR), as calculated per ANSI S3.19-1974, of the medical balloon-based earplug was 10 dB while that of a Peltor H10A earmuff was 24 dB. Although the medical balloon-based earplug did not prove to be a high attenuation-hearing protector, it produced a unique flat attenuation across the frequency spectrum, as compared to the typical increasing-with-frequency attenuation, pointing to its potential utility for applications wherein the pitch perception of sound is important.
Filename Size Approximate Download Time (Hours:Minutes:Seconds)
28.8 Modem 56K Modem ISDN (64 Kb) ISDN (128 Kb) Higher-speed Access Lee_K_D_2011.pdf 3.48 Mb 00:16:07 00:08:17 00:07:15 00:03:37 00:00:18 Lee_K_D_2011_Copyright.pdf 699.46 Kb 00:03:14 00:01:39 00:01:27 00:00:43 00:00:03
If you have questions or technical problems, please Contact DLA.